- Case report
- Open Access
A case of transvaginal NOTES partial gastrectomy using new techniques and devices
Surgical Case Reports volume 1, Article number: 96 (2015)
The latest technique of transvaginal NOTES partial gastrectomy is described in detail. The procedure involves new “over-tube steering” technique and usage of two newly developed endoscopic accessories. The technique is feasible, safe, and practical, since all devices used in the case are off-the-shelf products.
Ten years have passed since the original concept of NOTES (Natural Orifice Translumenal Endoscopic Surgery) was proposed by Kalloo et al. . The concept of intra-abdominal surgery using flexible endoscopy without causing major destruction of the abdominal wall was extremely innovative at that time; therefore, numerous preclinical feasibility/safety studies have been conducted by surgeons and gastroenterologists [2–6]. Although less postoperative pain and better cosmesis have been sporadically reported, its true less invasiveness has not been clearly described. The clinical application of NOTES thus has still remained in “exploratory” phase in limited advanced institutions, and various problems including medical, technical, ethical, and economical issues remain unsolved [3, 7, 8]. Is clinical NOTES already dead or alive? The authors believe that only a steady continuation of preclinical/clinical researches may address this question.
The authors formed a multi-disciplinary NOTES team with surgeons, gastroenterologists, and gynecologists and started conjoined preclinical research in 2006 [4, 6]. Our team then successfully performed the world’s first human transvaginal NOTES partial gastrectomy in a hybrid manner in 2008 . To date, we have continuously improved our surgical techniques using commercially available instruments and newly developed devices . In this communication, we describe our latest technique of transvaginal hybrid NOTES gastrectomy in a case with gastric gastrointestinal stromal tumor (GIST), using new techniques and newly developed endoscopic devices.
A 38-year-old female patient with an incidentally found gastric submucosal tumor was referred to our hospital for the consultation of the possibility of NOTES gastrectomy. The patient herself wished to undergo NOTES as a surgical alternative, due to possible less pain and better cosmesis. Her body weight was 41 kg and height was 155 cm. The patient was asymptomatic and otherwise healthy without any gastrointestinal disorders. The lesion, 2.5 cm in size, was demonstrated on the posterior wall of the gastric fornix by preoperative abdominal computed tomography. Endoscopy revealed a SMT with intra-gastric growth, with the distance of 2.0 cm below the esophago-gastric junction (Fig. 1). Biopsy was not attempted due to the size and location of the lesion. The patient experienced one childbirth with normal transvaginal delivery. After intensive discussion among the team, transvaginal NOTES partial gastrectomy was indicated as a surgical alternative. We informed the patient of its experimental nature, our previous clinical experience, the procedure’s short- and long-term outcomes, the possible risks/complications, and the details of the procedure itself. Surgery was conducted under the protocol approved by the University’s Institutional Review Board (IRB Folio No. 08063). Table 1 demonstrates our inclusion/exclusion criteria for transvaginal NOTES gastrectomy, which was approved by the University’s Institutional Review Board.
The routine part of our transvaginal NOTES technique has been described elsewhere [6, 7]. Briefly, after placing two laparoscopic ports at the umbilicus (5 mm) and the left mid-abdomen (3 mm), the posterior vaginal sac was exposed and punctured under laparoscopic guidance. This entry site was secured with an extra-long endoscopic over-tube, and a standard flexible gastrointestinal endoscope was advanced into the abdomen. The peri-gastric ligaments were dissected with energy devices passed through the endoscope to mobilize the involved portion of the stomach. The endoscope was then replaced with a linear stapling device. EGD was advanced trans-orally into the stomach to calibrate the EGJ and to obtain adequate surgical margin during stapling (so-called gastric calibration). Stapled partial gastrectomy was accomplished under laparoscopic monitoring. The specimen was isolated in a plastic bag and was delivered via the vagina. The vaginal entry site was closed with sutures under direct vision.
In this particular patient, we employed new techniques and newly developed accessory devices.
New technique: over-tube steering using loop-retracting device
The transvaginal over-tube was secured with a thin loop-retracting device (Mini Loop Retractor, Hakko, Nagano, Japan), which was introduced via the umbilical incision under retroflexed flexible endoscopic guidance. This technique enabled intentional “steering” of long and flexible over-tube in the abdominal cavity and greatly enhanced its stability while controlling the tip of the flexible endoscope (Fig. 2).
New device 1: leak-proof adapter for endoscopic over-tube
Currently available over-tube has been designed to be used as a “guide” for flexible endoscope, and is not suitable to maintain pneumoperitoneum. Its distal orifice is not leak-proof; therefore, carbon dioxide, once fed through the trans-abdominal trocar, continuously leaks from the space between the over-tube and the endoscope. To solve this problem, we have developed a reinforced anti-leak adapter, and it was commercialized as an endoscopic accessory (Leak Cutter, Top Corporation, Tokyo, Japan) in 2012. With this simple device, pneumoperitoneum was successfully established and maintained all through the procedure without causing any major gas leakage (Fig. 3).
New device 2: flexible irrigation/suction catheter
This newly developed device (Endoshower, Yamashina Seiki Co., Ltd., Siga, Japan) is a flexible catheter which passes the standard biopsy channel of current flexible endoscopes. Endoshower enables “laparoscopy-like” irrigation and suction under sterile condition. With this device, the abdomen was diffusely irrigated, and the fluid in the sub-diaphragmatic space was safely drained without causing any mis-suction and injuries on the splenic capsule (Fig. 4).
The procedure was successfully completed without any complications. No additional trocar was required, and no trans-abdominal energy device was used. The operative time was 260 min, and the blood loss was 20 ml. The patient showed uneventful and rapid postoperative recovery without any analgesic. She discharged from the hospital without any abdominal/gynecological complaints on the fifth postoperative day, which was determined by the IRB-approved clinical protocol. The postoperative pathology revealed leiomyoma of the stomach, 28 × 15 × 10 mm in size. Both umbilical and mid-abdominal scars are now well consolidated and virtually invisible.
One of the technical challenges in human NOTES is how to control and stabilize flexible endoscope in the abdominal cavity [2, 3, 7, 8]. Various commercially available/non-available over-tubes, i.e., platforms, have been attempted; however, virtually none of them have archived true clinical success [7, 8]. Our “over-tube steering technique” using loop-retracting device is simple, effective, and inexpensive; however, its true advantage in reducing operating time/blood loss was not shown in this single case report. This technique is applicable to any type of NOTES and practical since it is compatible with any kind of off-the-shelf endoscopic over-tubes. Further accumulation of clinical cases is definitely needed, to establish this technique as an attractive surgical option. The authors believe that future platform for pure NOTES should have at least “anchoring” mechanism that anchors the platform onto the abdominal lining.
How to maintain pneumoperitoneum is another fundamental but difficult problem in NOTES [5, 11]. Since commercially available endoscopic over-tubes are not designed to maintain a steady-pressure environment, they are not as leak-proof as laparoscopic trocars. Our newly developed “Leak Cutter,” now available in Japanese market as a reinforced anti-leak adapter for routine endoscopy, works perfect to prevent gas leakage during the case. The device has a Luer lock-type stopcock for emergent evacuation in case of over-insufflation. This stopcock can also be used as an insufflation inlet in future pure NOTES, where trans-abdominal insufflation via laparoscopic trocar will no longer be available.
The intra-abdominal irrigation via current flexible endoscope is feasible; however, it is technically demanding due to its one-direction spray from the tip of the endoscope. The use of sterile saline solution via built-in air/water channel requires specific and cumbersome preparation and setting. The suction from the tip of the endoscope is further challenging, since conventional technique requires “diving” of the endoscope beneath the fluid. While diving the endoscope, the visualization is temporally lost. This is acceptable in the gut lumen where no other organs exist; however, this is not acceptable in the abdominal cavity where blind mis-suction may lead to serious organ damage. Our new product, Endoshower, is one of the solutions. This device, 2.6 mm in outer diameter, has 24 side holes of 0.4 mm in size at its tip. With this device, sterile irrigation and safe suction under endoscopic visual control were feasible in the present case.
Transvaginal hybrid NOTES partial gastrectomy has been refined with several new ideas and devices. Clinical NOTES is still in the early developmental stage. Continuous efforts of medical device development and further accumulation of clinical experiences are necessary to make NOTES a “reality” rather than a “dream.”
The patient has provided permission to publish these features of her case, and the identity of the patient has been protected.
Kalloo AN, Singh VK, Jagannath SB, Niiyama H, Hill SL, Vaughn CA, et al. Flexible transgastric peritoneoscopy: a novel approach to diagnostic and therapeutic interventions in the peritoneal cavity. Gastrointest Endosc. 2004;60:114–7.
ASGE/SAGES Working Group on natural orifice translumenal endoscopic surgery white paper October 2005. Gastrointest Endosc. 2006;63:199–203.
Rattner DW, Hawes R, Schwaitzberg S, Kochman M, Swanstrom L. The Second SAGES/ASGE White Paper on natural orifice transluminal endoscopic surgery: 5 years of progress. Surg Endosc. 2011;25:2441–8.
Nakajima K, Souma Y, Takahashi T, Yamasaki M, Miyazaki Y, Mori M, et al. Anatomical measurements to optimize instrumentation for transvaginal surgery. Surg Endosc. 2013;27:2052–7.
Nakajima K, Nishida T, Milsom JW, Takahashi T, Souma Y, Miyazaki Y, et al. Current limitations in endoscopic CO2 insufflation for NOTES: flow and pressure study. Gastrointest Endosc. 2010;72:1036–42.
Nakajima K, Moon JH, Tsutsui S, Miyazaki Y, Yamasaki M, Yamada T, et al. Esophageal submucosal dissection under steady pressure automatically controlled endoscopy (SPACE): a randomized preclinical trial. Surg Endosc. 2012;44:1139–48.
Kitano S, Tajiri H, Yasuda K, Ikeda K, Sumiyama K, Nakajima K, et al. Japan NOTES working group for white paper constitution. Current status and activity regarding natural orifice translumenal endoscopic surgery (NOTES) in Japan. Asian J Endosc Surg. 2008;1:7–10.
Flora ED, Wilson TG, Martin IJ, O’Rourke NA, Maddern GJ. A review of natural orifice translumenal endoscopic surgery (NOTES) for intra-abdominal surgery: experimental models, techniques, and applicability to the clinical setting. Ann Surg. 2008;247:583–602.
Nakajima K, Nishida T, Takahashi T, Souma Y, Hara J, Yamada T, et al. Partial gastrectomy using natural orifice translumenal endoscopic surgery (NOTES) for gastric submucosal tumors: early experience in humans. Surg Endosc. 2009;23:2650–5.
Hirota M, Nakajima K, Hara J, Takahashi T, Yamasaki M, Tsutsui T, et al. A case of simultaneous transvaginal NOTES gastrectomy and vaginal hysterectomy in a patient with gastric submucosal tumor and uterine prolapse. Asian J Endosc Surg. 2012;5:168–71.
von Delius S, Huber W, Feussner H, Wilhelm D, Karagianni A, Henke J, et al. Effect of pneumoperitoneum on hemodynamics and inspiratory pressures during natural orifice transluminal endoscopic surgery (NOTES): an experimental, controlled study in an acute porcine model. Endoscopy. 2007;39:854–61.
The authors declare that they have no competing interests.
KN, TT, and SH carried out the surgical procedure in the present study. All authors participated in the management of the patient in this case report. All authors read and approved the final manuscript.
About this article
Cite this article
Higashi, S., Nakajima, K., Miyazaki, Y. et al. A case of transvaginal NOTES partial gastrectomy using new techniques and devices. surg case rep 1, 96 (2015) doi:10.1186/s40792-015-0093-6
- Natural orifice translumenal endoscopic surgery