- Case Report
- Open Access
Bronchial artery aneurysm suggested to be caused by metalic tracheal stent migration
© The Author(s). 2016
- Received: 9 July 2016
- Accepted: 22 October 2016
- Published: 5 November 2016
Occurrence of bronchial artery aneurysm is rare, and it has been detected in less than 1 % of all selective bronchial arteriography cases. Here, we present a case of a bronchial artery aneurysm caused by a tracheal stent migration. A 59-year-old man was operated on for esophageal cancer, where an esophageal-tracheal fistula occurred 1 week after operation. Surgical repair of the esophageal-tracheal fistula was performed using a muscle flap, but this not results in fistula closure. Consequently, a self-expanding covered metallic tracheal stent was implanted for rescue, and this resulted in fistula closure. After 1 year, there was frequent hemoptysis caused by migration of the stent. He was referred to our hospital where removal of the stent was planned. A sudden occurrence of massive bleeding from trachea occurred, and extracorporeal membrane oxygenation (ECMO) was used. Although removal of tracheal stent was performed successfully, the patient subsequently died from multi-organ failure. Post-mortem autopsy revealed that the massive bleeding is originated from the rupture of a bronchial artery aneurysm.
- Bronchial Artery
- Metallic Stents
- Tracheal Stenosis
- Stent Removal
- Latissimus Dorsi Muscle
Self-expanding metallic tracheal stents have been used for benign or malignant tracheal stenosis and tracheal fistulas. Although the implantation of these stents are considered as a relatively safe treatment, severe complications such as bleeding and stent migration can sometimes occur. Here, we report a case of massive bleeding originating from the rupture of a bronchial artery aneurysm suggested to be caused by a metalic tracheal stent migration.
Recently, posterior mediastinal reconstruction is increasingly used as the reconstructive route for patients undergoing an esophagectomy for esophagus cancer. Occurrence of gastric tube-bronchial fistula after esophagectomy is a rare but potentially life-threatening complication that is specific to posterior mediastinal reconstruction, and incidence of this complication has been reported to be around 3 % . This complication can result from perioperative tracheal injury, ischemic lesions due to devascularization of the trachea and main bronchus by extensive lymph node dissection, or leakage of the esophagointestinal anastomosis, and it may present early after esophagectomy or relatively late in the follow-up. Management strategy is dependent on the fistula site and size and the patient’s characteristics and general condition. Although various management approaches have been proposed, the optimal strategy remains controversial. Although some authors have reported successful cases of stent insertion for the fistula , appropriate surgical repair is required in many cases [3, 4]. Nonsurgical management, such as airway stents, can be considered for a small fistula with minimal clinical symptoms or patients who are unable to go through surgery. Silicone and self-expanding covered metallic stents have been used in patients with fistula after esophageal surgery. Insertion of a silicone stent such as a Dumon stent requires a rigid bronchoscope and specialized technical skills, and general anesthesia is usually required. In contrast, a self-expanding metallic stent is easy to insert through a delivery catheter using a flexible bronchoscope under local anesthesia. The most common complications of these stents are hemoptysis, stent migration, and stent granulation. In many reports, self-expanding metallic stents have a lower rate of migration and hemoptysis, but a higher rate of granulation as compared with silicone stents [5–7]. There are also several additional complications including stent fracture, trachea/broncho-esophageal fistula , broncho-vascular fistula , and difficulty in removal. Removal of metallic stents can also result in serious complications, including mucosal tears and severe bleeding. Consequent to these complications, the US Food and Drug Administration recommends that metallic stents should only be used in patients with benign airway disorders after thoroughly exploring all other treatment options.
In our patient, we initially suspected that the hemoptysis was caused by tracheal mucosa damage, but bronchial arteriography and autopsy revealed that the bleeding was caused by perforation of a bronchial aneurysm. Little is known about the cause of bronchial artery aneurysm; however, it is regarded to be associated with increased bronchial artery blood flow or weakening or injury of the vessel walls . While congenital disease cannot be excluded in our case, weakening of the vessel walls by chronic irritation and inflammation from the stent is considered as the cause of the aneurysm. It is possible that the aneurysm might have been prevented if stent removal was performed early after the second reconstruction surgery. However, as previously mentioned, removal of a metallic stent is an inherently risky procedure, and therefore, stent removal is still subject to controversy.
We present a case of bronchial artery aneurysm caused by tracheal stent migration. Bronchial artery aneurysm is rare but serious complication in patients with using metallic tracheal stent. The indication of the metallic tracheal stent insertion should be made sufficiently carefully for benign airway disorders such as our case.
KS contributed to the writing of the manuscript. HS, YK, TM, NH, and TK supervised the study. SF, TF, KO, NK, and YI served as attending physicians of the presented patient. All authors read and approved the final manuscript.
The authors declare that they have no competing interests.
Consent for publication
Written informed consent was obtained from the patient’s bereaved family for publication of this case report and any accompanying images. A copy of the written consent is available for review by the Editor-in-Chief of this journal.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.
- Bartels HE, Stein HJ, Siewert RJ. Tracheobronchial lesions following oesophagectomy. Prevalence, predisposing factors and outcome. Br J Surg. 1998;85:403–6.View ArticlePubMedGoogle Scholar
- Schweigert M, Dubecz A, Beron M, Muschweck H, Stein HL. Management of anastomotic leakage-induced tracheobronchial fistula following oesophagectomy: the role of endoscopic stent insertion. Eur J Cardiothorac Surg. 2012;41:e74–80.View ArticlePubMedGoogle Scholar
- Shen KR, Allen MS, Cassivi SD, Nichols 3rd FC, Wigle DA, Harmsen WS, et al. Surgical management of acquired nonmalignant tracheoesophageal and bronchoesophageal fistulae. Ann Thorac Surg. 2010;90:914–9.View ArticlePubMedGoogle Scholar
- Buskens CJ, Hulscher JB, Fockens P, Obertop H, van Lanschot JJ. Benign trachea-neo-esophageal fistulas after subtotal esophagectomy. Ann Thorac Surg. 2001;72:221–4.View ArticlePubMedGoogle Scholar
- Martinez-Ballarin JI, Diaz-Jimenez JP, Castro MJ, Moya JA. Silicone stents in the management of benign tracheobronchial stenosis. Tolerance and early results in 63 patients. Chest. 1996;109:626–9.View ArticlePubMedGoogle Scholar
- Mitsuoka M, Sakuragi T, Itoh T. Clinical benefits and complications of Dumon stent insertion for the treatment of severe central airway stenosis or airway fistula. Gen Thorac Cardiovasc Surg. 2007;55:275–80.View ArticlePubMedGoogle Scholar
- McGrath EE, Warriner D, Anderson P. The insertion of self expanding metal stents with flexible bronchoscopy under sedation for malignant tracheobronchial stenosis: a single-center retrospective analysis. Arch Bronconeumol. 2012;48:43–8.Google Scholar
- Gaissert HA, Grillo HC, Wright CD, Donahue DM, Wain JC, Mathisen DJ. Complication of benign tracheobronchial strictures by self-expanding metal stents. J Thorac Cardiovasc Surg. 2003;126:744–7.View ArticlePubMedGoogle Scholar
- Choudhary C, Bandyopadhyay D, Salman R, Gildea T, Mehta A. Broncho-vascular fistulas from self-expanding metallic stents: a retrospective case review. Ann Thorac Medicine. 2013;8:116–20.View ArticleGoogle Scholar
- Lin J, Wood DE. Bronchial artery aneurysm refractory to transcatheter embolization. Ann Thorac Surg. 2008;86:306–8.View ArticlePubMedGoogle Scholar